- Empower 3
- Agilent 6890 GC
The following GC Inject error codes may appear in the Message Center table as
"Inject Error - X", where X may be one of the following numbers. The list below
contains interpretations (where available) of the codes.
0 No error.
100 Tray z axis failed to find home.
101 Tray Radius failed to find home.
102 Tray Theta failed to find home.
103 There was a bottle in the gripper.
104 There was no bottle in the gripper.
105 A stop key was pressed.
106 The selected injector is not present.
107 The tray is not present.
108 The Bar Code Reader is not present.
110 Tower/Controller communications error. No flashes on tower.
111 2 flashes on front tower.
112 3 flashes on front tower.
113 3 flashes on front tower.
114 5 flashes on front tower.
116 Front door was open.
117 Front tower was reset.
120 Tower/Controller communications error. No flashes on tower.
121 2 flashes on back tower.
122 3 flashes on back tower.
123 4 flashes on back tower.
124 5 flashes on back tower.
126 Back door was open.
127 Back tower was reset.
129 BCR driver detected an overload. Possible short in connector.
130 BCR driver detected a stalled mixer.
131 A bar code was not read.
132 HCBR has failed.
133 Out of range index to NAME_TABLE.
134 NAME_TABLE[Index] is empty.
135 Illegal name. NAME_TABLE[Index] was not changed.
136 Illegal location. NAME_TABLE[Index] was not changed.
137 Referenced name is not in the NAME_TABLE.
138 Invalid parameter sent to a command.
139 Comma needed.
140 Semicolon needed.
141 Numeric overflow > 231.
142 Range error in SETPTS command.
143 Error in SETPTS command.
144 Attempt to access an invalid tray position.
145 Attempt to access an invalid injector turret position.