- Degasser hardware fault.
- Resetting binary solvent manager clears error for short time, but it returns.
- ACQUITY Binary Solvent Manager (BSM)
End of degasser pump exhaust tube below level of solvent in waste jug.
FIX or WORKAROUND
- Pull exhaust tubing up so the end is not below the level of solvent in the waste jug.
- Reset BSM.
This fault can also be as a result of a defective degasser pump (part number: 700001352).