- After processing, the standard peak does not have any amounts or a calibration curve.
The spelling of the peak name in the amounts table does not match the spelling of the peak name in the components table. The mismatch could be due to spelling or capitalization.
FIX or WORKAROUND
- Correct the spelling in the components table and reprocess the data.