- Sample set is sometimes interrupted, and an instrument failure is recorded in message center
- Event logs give no clues
- Instrument server log shows “EndOfRun Controller Msg (Event Only - WaitProxiesComplete): 6890@3, Code 24 (not successful)”.
- Empower 3 Feature Release 2
- Avocent Driver: 220.127.116.11
- Agilent GC 6890
Unknown (defect PCS 51486 and CRI-296)
FIX or WORKAROUND
As a workaround, change the connectivity of the Agilent GC 6890 from Avocent to the PC COM port.